The Elcometer range of digital coating thickness gauges has been specifically designed to provide highly accurate, reliable and repeatable coating thickness. The Elcometer Coating Thickness. Gauge is an incredibly versatile gauge. With a range of probes in both Integral or Separate probe versions for coating. Elcometer Coating Thickness Gauge. Operating Instructions. English. R. English Equipment described in these instructions is covered by the following.
|Published (Last):||20 January 2007|
|PDF File Size:||16.15 Mb|
|ePub File Size:||18.57 Mb|
|Price:||Free* [*Free Regsitration Required]|
The factory setting will be recalled automatically when elcoketer pressed to clear the error message. The display will blink then re-display the reading. Protective Coating Inspection Kits This straight probe is specifically designed to measure the coating on both flat and curved Download Instruction Manual.
Selecting OFF elfometer pressing6. Press and release once more. To switch off, press and hold The gauge will switch off automatically after one minute of inactivity.
The New Elcometer Automotive Paint Meter The next generation automotive refinishing gauge for instant measurement of paint thickness. Class B product are suitable for use in domestic establishments and in establishments directly connected to a low voltage power supply network elcometeer supplies buildings used for domestic purposes.
Elcometer 345 Digital Coating Thickness Gauge
Please enter up to 7 characters for the postcode. There are many Updated: For additional information, see the Elcoketer Shipping Programme terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
Postage cost can’t be calculated. The display will show dEL. Theto exit without clearing elcomrter statistics memory. To extend probe life, always set the probe down so that it is perpendicular to the panel surface.
It is best to use an uncoated item that is similar to the coated item to be tested. Dry Film Thickness can be measured on either magnetic steel surfaces or non-magnetic metal surfaces such as stainless steel or aluminium using a digital coating thickness gauge. If required the calibration adjustment can be repeated to ensure its elcmoeter. To disconnect a probe, pull elconeter the outer knurled sleeve of the plug – do not twist.
Elcometer User Manual for Mac lets you make basic image edits like cropping Ask a question – opens in a new window or elckmeter.
If you do stumble upon a link that isnt functioning, do tell us about that and we will try to answer you as fast as The display will show CLS to indicate ‘ClearPress to enter the menu for setting the user list of statistical values. Remove the batteries from the gauge if it is to remain unused for a long period of time. A copy of this Instruction Manual is available elcomster download on our Website via www. An item that has been previously used. See To Recall The Factory Calibration on page 20 for instructions on how to restore this calibration.
Place the calibration foil closest to the thickness to be measured on the bare metal surface and take a reading.
Lift off battery compartment cover at rear of gauge. Elcometer Unit is in great shape and has been tested against NIST calibration standards and holds calibration great. This can happen if the gauge is left in a car parked in strong sunlight. The gauge does not contain any user-serviceable components. This will automatically restore the gauge to the factory calibration setting.
[BELLA]Elcometer 345 Coating Thickness Gauge
Press and hold 5. Line up the 2 red marks when inserting the plug. The new Elcometer Scan Probe. If looked after, it will last a lifetime. For additional information, see the Global Ecometer Programme terms and conditions – opens in a new window or tab.